#!/usr/bin/env python
# -*- coding: utf-8 -*-
"""
Classes related to instrument responses.

:copyright:
    Lion Krischer (krischer@geophysik.uni-muenchen.de), 2013
:license:
    GNU Lesser General Public License, Version 3
    (https://www.gnu.org/copyleft/lesser.html)
"""
import copy
import ctypes as C  # NOQA
from collections import defaultdict
from copy import deepcopy
import itertools
from math import pi
import warnings

import numpy as np

from .. import compatibility
from obspy.core.util.base import ComparingObject
from obspy.core.util.deprecation_helpers import ObsPyDeprecationWarning
from obspy.core.util.obspy_types import (ComplexWithUncertainties,
                                         FloatWithUncertainties,
                                         FloatWithUncertaintiesAndUnit,
                                         ObsPyException,
                                         ZeroSamplingRate)

from .util import Angle, Frequency


class ResponseStage(ComparingObject):
    """
    From the StationXML Definition:
        This complex type represents channel response and covers SEED
        blockettes 53 to 56.
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 resource_id=None, resource_id2=None, name=None,
                 input_units_description=None,
                 output_units_description=None, description=None,
                 decimation_input_sample_rate=None, decimation_factor=None,
                 decimation_offset=None, decimation_delay=None,
                 decimation_correction=None):
        """
        :type stage_sequence_number: int
        :param stage_sequence_number: Stage sequence number, greater or equal
            to zero. This is used in all the response SEED blockettes.
        :type stage_gain: float
        :param stage_gain: Value of stage gain.
        :type stage_gain_frequency: float
        :param stage_gain_frequency: Frequency of stage gain.
        :type input_units: str
        :param input_units: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Name of units, e.g. "M/S", "V", "PA".
        :type output_units: str
        :param output_units: The units of the data as output from the
            perspective of data acquisition. These would be the units of the
            data prior to correcting for this response.
            Name of units, e.g. "M/S", "V", "PA".
        :type resource_id: str
        :param resource_id: This field contains a string that should serve as a
            unique resource identifier. This identifier can be interpreted
            differently depending on the data center/software that generated
            the document. Also, we recommend to use something like
            GENERATOR:Meaningful ID. As a common behavior equipment with the
            same ID should contains the same information/be derived from the
            same base instruments.
        :type resource_id2: str
        :param resource_id2: This field contains a string that should serve as
            a unique resource identifier. Resource identifier of the subgroup
            of the response stage that varies across different response stage
            types (e.g. the poles and zeros part or the FIR part).
        :type name: str
        :param name: A name given to the filter stage.
        :type input_units_description: str, optional
        :param input_units_description: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :type output_units_description: str, optional
        :param output_units_description: The units of the data as output from
            the perspective of data acquisition. These would be the units of
            the data prior to correcting for this response.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :type description: str, optional
        :param description: A short description of of the filter.
        :type decimation_input_sample_rate:  float, optional
        :param decimation_input_sample_rate: The sampling rate before the
            decimation in samples per second.
        :type decimation_factor: int, optional
        :param decimation_factor: The applied decimation factor.
        :type decimation_offset: int, optional
        :param decimation_offset: The sample chosen for use. 0 denotes the
            first sample, 1 the second, and so forth.
        :type decimation_delay: float, optional
        :param decimation_delay: The estimated pure delay from the decimation.
        :type decimation_correction: float, optional
        :param decimation_correction: The time shift applied to correct for the
            delay at this stage.

        .. note::
            The stage gain (or stage sensitivity) is the gain at the stage of
            the encapsulating response element and corresponds to SEED
            blockette 58. In the SEED convention, stage 0 gain represents the
            overall sensitivity of the channel.  In this schema, stage 0 gains
            are allowed but are considered deprecated.  Overall sensitivity
            should be specified in the InstrumentSensitivity element.
        """
        self.stage_sequence_number = stage_sequence_number
        self.input_units = input_units
        self.output_units = output_units
        self.input_units_description = input_units_description
        self.output_units_description = output_units_description
        self.resource_id = resource_id
        self.resource_id2 = resource_id2
        self.stage_gain = stage_gain
        self.stage_gain_frequency = stage_gain_frequency
        self.name = name
        self.description = description
        self.decimation_input_sample_rate = \
            Frequency(decimation_input_sample_rate) \
            if decimation_input_sample_rate is not None else None
        self.decimation_factor = decimation_factor
        self.decimation_offset = decimation_offset
        self.decimation_delay = \
            FloatWithUncertaintiesAndUnit(decimation_delay) \
            if decimation_delay is not None else None
        self.decimation_correction = \
            FloatWithUncertaintiesAndUnit(decimation_correction) \
            if decimation_correction is not None else None

    def __str__(self):
        ret = (
            "Response type: {response_type}, Stage Sequence Number: "
            "{response_stage}\n"
            "{name_desc}"
            "{resource_id}"
            "\tFrom {input_units}{input_desc} to {output_units}{output_desc}\n"
            "\tStage gain: {gain}, defined at {gain_freq} Hz\n"
            "{decimation}").format(
            response_type=self.__class__.__name__,
            response_stage=self.stage_sequence_number,
            gain=self.stage_gain if self.stage_gain is not None else "UNKNOWN",
            gain_freq=("%.2f" % self.stage_gain_frequency) if
            self.stage_gain_frequency is not None else "UNKNOWN",
            name_desc="\t%s %s\n" % (
                self.name, "(%s)" % self.description
                if self.description else "") if self.name else "",
            resource_id=("\tResource Id: %s" % self.resource_id
                         if self.resource_id else ""),
            input_units=self.input_units if self.input_units else "UNKNOWN",
            input_desc=(" (%s)" % self.input_units_description
                        if self.input_units_description else ""),
            output_units=self.output_units if self.output_units else "UNKNOWN",
            output_desc=(" (%s)" % self.output_units_description
                         if self.output_units_description else ""),
            decimation=(
                "\tDecimation:\n\t\tInput Sample Rate: %.2f Hz\n\t\t"
                "Decimation Factor: %i\n\t\tDecimation Offset: %i\n\t\t"
                "Decimation Delay: %.2f\n\t\tDecimation Correction: %.2f" % (
                    self.decimation_input_sample_rate, self.decimation_factor,
                    self.decimation_offset, self.decimation_delay,
                    self.decimation_correction)
                if self.decimation_input_sample_rate is not None else ""))
        return ret.strip()

    def _repr_pretty_(self, p, cycle):
        p.text(str(self))


class PolesZerosResponseStage(ResponseStage):
    """
    From the StationXML Definition:
        Response: complex poles and zeros. Corresponds to SEED blockette 53.

    The response stage is used for the analog stages of the filter system and
    the description of infinite impulse response (IIR) digital filters.

    Has all the arguments of the parent class
    :class:`~obspy.core.inventory.response.ResponseStage` and the following:

    :type pz_transfer_function_type: str
    :param pz_transfer_function_type: A string describing the type of transfer
        function. Can be one of:

        * ``LAPLACE (RADIANS/SECOND)``
        * ``LAPLACE (HERTZ)``
        * ``DIGITAL (Z-TRANSFORM)``

        The function tries to match inputs to one of three types if it can.
    :type normalization_frequency: float
    :param normalization_frequency: The frequency at which the normalization
        factor is normalized.
    :type zeros: list[complex]
    :param zeros: All zeros of the stage.
    :type poles: list[complex]
    :param poles: All poles of the stage.
    :type normalization_factor: float, optional
    :param normalization_factor:
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 pz_transfer_function_type,
                 normalization_frequency, zeros, poles,
                 normalization_factor=1.0, resource_id=None, resource_id2=None,
                 name=None, input_units_description=None,
                 output_units_description=None, description=None,
                 decimation_input_sample_rate=None, decimation_factor=None,
                 decimation_offset=None, decimation_delay=None,
                 decimation_correction=None):
        # Set the Poles and Zeros specific attributes. Special cases are
        # handled by properties.
        self.pz_transfer_function_type = pz_transfer_function_type
        self.normalization_frequency = normalization_frequency
        self.normalization_factor = float(normalization_factor)
        self.zeros = zeros
        self.poles = poles
        super(PolesZerosResponseStage, self).__init__(
            stage_sequence_number=stage_sequence_number,
            input_units=input_units,
            output_units=output_units,
            input_units_description=input_units_description,
            output_units_description=output_units_description,
            resource_id=resource_id, resource_id2=resource_id2,
            stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency, name=name,
            description=description,
            decimation_input_sample_rate=decimation_input_sample_rate,
            decimation_factor=decimation_factor,
            decimation_offset=decimation_offset,
            decimation_delay=decimation_delay,
            decimation_correction=decimation_correction)

    def __str__(self):
        ret = super(PolesZerosResponseStage, self).__str__()
        ret += (
            "\n"
            "\tTransfer function type: {transfer_fct_type}\n"
            "\tNormalization factor: {norm_fact:g}, "
            "Normalization frequency: {norm_freq:.2f} Hz\n"
            "\tPoles: {poles}\n"
            "\tZeros: {zeros}").format(
            transfer_fct_type=self.pz_transfer_function_type,
            norm_fact=self.normalization_factor,
            norm_freq=self.normalization_frequency,
            poles=", ".join(map(str, self.poles)),
            zeros=", ".join(map(str, self.zeros)))
        return ret

    def _repr_pretty_(self, p, cycle):
        p.text(str(self))

    @property
    def zeros(self):
        return self._zeros

    @zeros.setter
    def zeros(self, value):
        value = list(value)
        for i, x in enumerate(value):
            if not isinstance(x, ComplexWithUncertainties):
                value[i] = ComplexWithUncertainties(x)
        self._zeros = value

    @property
    def poles(self):
        return self._poles

    @poles.setter
    def poles(self, value):
        value = list(value)
        for i, x in enumerate(value):
            if not isinstance(x, ComplexWithUncertainties):
                value[i] = ComplexWithUncertainties(x)
        self._poles = value

    @property
    def normalization_frequency(self):
        return self._normalization_frequency

    @normalization_frequency.setter
    def normalization_frequency(self, value):
        self._normalization_frequency = Frequency(value)

    @property
    def pz_transfer_function_type(self):
        return self._pz_transfer_function_type

    @pz_transfer_function_type.setter
    def pz_transfer_function_type(self, value):
        """
        Setter for the transfer function type.

        Rather permissive but should make it less awkward to use.
        """
        msg = ("'%s' is not a valid value for 'pz_transfer_function_type'. "
               "Valid one are:\n"
               "\tLAPLACE (RADIANS/SECOND)\n"
               "\tLAPLACE (HERTZ)\n"
               "\tDIGITAL (Z-TRANSFORM)") % value
        value = value.lower()
        if "laplace" in value:
            if "radian" in value:
                self._pz_transfer_function_type = "LAPLACE (RADIANS/SECOND)"
            elif "hertz" in value or "hz" in value:
                self._pz_transfer_function_type = "LAPLACE (HERTZ)"
            else:
                raise ValueError(msg)
        elif "digital" in value:
            self._pz_transfer_function_type = "DIGITAL (Z-TRANSFORM)"
        else:
            raise ValueError(msg)


class CoefficientsTypeResponseStage(ResponseStage):
    """
    This response type can describe coefficients for FIR filters. Laplace
    transforms and IIR filters can also be expressed using this type but should
    rather be described using the PolesZerosResponseStage class. Effectively
    corresponds to SEED blockette 54.

    Has all the arguments of the parent class
    :class:`~obspy.core.inventory.response.ResponseStage` and the following:

    :type cf_transfer_function_type: str
    :param cf_transfer_function_type: A string describing the type of transfer
        function. Can be one of:

        * ``ANALOG (RADIANS/SECOND)``
        * ``ANALOG (HERTZ)``
        * ``DIGITAL``

        The function tries to match inputs to one of three types if it can.
    :type numerator: list of
        :class:`~obspy.core.inventory.response.CoefficientWithUncertainties`
    :param numerator: Numerator of the coefficient response stage.
    :type denominator: list of
        :class:`~obspy.core.inventory.response.CoefficientWithUncertainties`
    :param denominator: Denominator of the coefficient response stage.
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 cf_transfer_function_type, resource_id=None,
                 resource_id2=None, name=None, numerator=None,
                 denominator=None, input_units_description=None,
                 output_units_description=None, description=None,
                 decimation_input_sample_rate=None, decimation_factor=None,
                 decimation_offset=None, decimation_delay=None,
                 decimation_correction=None):
        # Set the Coefficients type specific attributes. Special cases are
        # handled by properties.
        self.cf_transfer_function_type = cf_transfer_function_type
        self.numerator = numerator
        self.denominator = denominator
        super(CoefficientsTypeResponseStage, self).__init__(
            stage_sequence_number=stage_sequence_number,
            input_units=input_units,
            output_units=output_units,
            input_units_description=input_units_description,
            output_units_description=output_units_description,
            resource_id=resource_id, resource_id2=resource_id2,
            stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency, name=name,
            description=description,
            decimation_input_sample_rate=decimation_input_sample_rate,
            decimation_factor=decimation_factor,
            decimation_offset=decimation_offset,
            decimation_delay=decimation_delay,
            decimation_correction=decimation_correction)

    def __str__(self):
        ret = super(CoefficientsTypeResponseStage, self).__str__()
        ret += (
            "\n"
            "\tTransfer function type: {transfer_fct_type}\n"
            "\tContains {num_count} numerators and {den_count} denominators")\
            .format(
                transfer_fct_type=self.cf_transfer_function_type,
                num_count=len(self.numerator), den_count=len(self.denominator))
        return ret

    def _repr_pretty_(self, p, cycle):
        p.text(str(self))

    @property
    def numerator(self):
        return self._numerator

    @numerator.setter
    def numerator(self, value):
        if value == []:
            self._numerator = []
            return
        value = list(value) if isinstance(
            value, compatibility.collections_abc.Iterable) else [value]
        if any(getattr(x, 'unit', None) is not None for x in value):
            msg = 'Setting Numerator/Denominator with a unit is deprecated.'
            warnings.warn(msg, ObsPyDeprecationWarning)
        for _i, x in enumerate(value):
            if not isinstance(x, CoefficientWithUncertainties):
                value[_i] = CoefficientWithUncertainties(x)
        self._numerator = value

    @property
    def denominator(self):
        return self._denominator

    @denominator.setter
    def denominator(self, value):
        if value == []:
            self._denominator = []
            return
        value = list(value) if isinstance(
            value, compatibility.collections_abc.Iterable) else [value]
        if any(getattr(x, 'unit', None) is not None for x in value):
            msg = 'Setting Numerator/Denominator with a unit is deprecated.'
            warnings.warn(msg, ObsPyDeprecationWarning)
        for _i, x in enumerate(value):
            if not isinstance(x, CoefficientWithUncertainties):
                value[_i] = CoefficientWithUncertainties(x)
        self._denominator = value

    @property
    def cf_transfer_function_type(self):
        return self._cf_transfer_function_type

    @cf_transfer_function_type.setter
    def cf_transfer_function_type(self, value):
        """
        Setter for the transfer function type.

        Rather permissive but should make it less awkward to use.
        """
        msg = ("'%s' is not a valid value for 'cf_transfer_function_type'. "
               "Valid one are:\n"
               "\tANALOG (RADIANS/SECOND)\n"
               "\tANALOG (HERTZ)\n"
               "\tDIGITAL") % value
        value = value.lower()
        if "analog" in value:
            if "rad" in value:
                self._cf_transfer_function_type = "ANALOG (RADIANS/SECOND)"
            elif "hertz" in value or "hz" in value:
                self._cf_transfer_function_type = "ANALOG (HERTZ)"
            else:
                raise ValueError(msg)
        elif "digital" in value:
            self._cf_transfer_function_type = "DIGITAL"
        else:
            raise ValueError(msg)


class ResponseListResponseStage(ResponseStage):
    """
    This response type gives a list of frequency, amplitude and phase value
    pairs. Effectively corresponds to SEED blockette 55.

    Has all the arguments of the parent class
    :class:`~obspy.core.inventory.response.ResponseStage` and the following:

    :type response_list_elements: list of
        :class:`~obspy.core.inventory.response.ResponseListElement`
    :param response_list_elements: A list of single discrete frequency,
        amplitude and phase response values.
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 resource_id=None, resource_id2=None, name=None,
                 response_list_elements=None,
                 input_units_description=None, output_units_description=None,
                 description=None, decimation_input_sample_rate=None,
                 decimation_factor=None, decimation_offset=None,
                 decimation_delay=None, decimation_correction=None):
        self.response_list_elements = response_list_elements or []
        super(ResponseListResponseStage, self).__init__(
            stage_sequence_number=stage_sequence_number,
            input_units=input_units,
            output_units=output_units,
            input_units_description=input_units_description,
            output_units_description=output_units_description,
            resource_id=resource_id, resource_id2=resource_id2,
            stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency, name=name,
            description=description,
            decimation_input_sample_rate=decimation_input_sample_rate,
            decimation_factor=decimation_factor,
            decimation_offset=decimation_offset,
            decimation_delay=decimation_delay,
            decimation_correction=decimation_correction)


class ResponseListElement(ComparingObject):
    """
    Describes the amplitude and phase response value for a discrete frequency
    value.
    """
    def __init__(self, frequency, amplitude, phase):
        """
        :type frequency: float
        :param frequency: The frequency for which the response is valid.
        :type amplitude: float
        :param amplitude: The value for the amplitude response at this
            frequency.
        :type phase: float
        :param phase: The value for the phase response at this frequency.
        """
        self.frequency = frequency
        self.amplitude = amplitude
        self.phase = phase

    @property
    def frequency(self):
        return self._frequency

    @frequency.setter
    def frequency(self, value):
        if not isinstance(value, Frequency):
            value = Frequency(value)
        self._frequency = value

    @property
    def amplitude(self):
        return self._amplitude

    @amplitude.setter
    def amplitude(self, value):
        if not isinstance(value, FloatWithUncertaintiesAndUnit):
            value = FloatWithUncertaintiesAndUnit(value)
        self._amplitude = value

    @property
    def phase(self):
        return self._phase

    @phase.setter
    def phase(self, value):
        if not isinstance(value, Angle):
            value = Angle(value)
        self._phase = value


class FIRResponseStage(ResponseStage):
    """
    From the StationXML Definition:
        Response: FIR filter. Corresponds to SEED blockette 61. FIR filters are
        also commonly documented using the CoefficientsType element.

    Has all the arguments of the parent class
    :class:`~obspy.core.inventory.response.ResponseStage` and the following:

    :type symmetry: str
    :param symmetry: A string describing the symmetry. Can be one of:

            * ``NONE``
            * ``EVEN``
            * ``ODD``

    :type coefficients: list[float]
    :param coefficients: List of FIR coefficients.
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 symmetry="NONE", resource_id=None, resource_id2=None,
                 name=None,
                 coefficients=None, input_units_description=None,
                 output_units_description=None, description=None,
                 decimation_input_sample_rate=None, decimation_factor=None,
                 decimation_offset=None, decimation_delay=None,
                 decimation_correction=None):
        self._symmetry = symmetry
        self.coefficients = coefficients or []
        super(FIRResponseStage, self).__init__(
            stage_sequence_number=stage_sequence_number,
            input_units=input_units,
            output_units=output_units,
            input_units_description=input_units_description,
            output_units_description=output_units_description,
            resource_id=resource_id, resource_id2=resource_id2,
            stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency, name=name,
            description=description,
            decimation_input_sample_rate=decimation_input_sample_rate,
            decimation_factor=decimation_factor,
            decimation_offset=decimation_offset,
            decimation_delay=decimation_delay,
            decimation_correction=decimation_correction)

    @property
    def symmetry(self):
        return self._symmetry

    @symmetry.setter
    def symmetry(self, value):
        value = str(value).upper()
        allowed = ("NONE", "EVEN", "ODD")
        if value not in allowed:
            msg = ("Value '%s' for FIR Response symmetry not allowed. "
                   "Possible values are: '%s'")
            msg = msg % (value, "', '".join(allowed))
            raise ValueError(msg)
        self._symmetry = value

    @property
    def coefficients(self):
        return self._coefficients

    @coefficients.setter
    def coefficients(self, value):
        new_values = []
        for x in value:
            if not isinstance(x, FilterCoefficient):
                x = FilterCoefficient(x)
            new_values.append(x)
        self._coefficients = new_values


class PolynomialResponseStage(ResponseStage):
    """
    From the StationXML Definition:
        Response: expressed as a polynomial (allows non-linear sensors to be
        described). Corresponds to SEED blockette 62. Can be used to describe a
        stage of acquisition or a complete system.

    Has all the arguments of the parent class
    :class:`~obspy.core.inventory.response.ResponseStage` and the following:

    :type approximation_type: str
    :param approximation_type: Approximation type. Currently restricted to
        'MACLAURIN' by StationXML definition.
    :type frequency_lower_bound: float
    :param frequency_lower_bound: Lower frequency bound.
    :type frequency_upper_bound: float
    :param frequency_upper_bound: Upper frequency bound.
    :type approximation_lower_bound: float
    :param approximation_lower_bound: Lower bound of approximation.
    :type approximation_upper_bound: float
    :param approximation_upper_bound: Upper bound of approximation.
    :type maximum_error: float
    :param maximum_error: Maximum error.
    :type coefficients: list[float]
    :param coefficients: List of polynomial coefficients.
    """
    def __init__(self, stage_sequence_number, stage_gain,
                 stage_gain_frequency, input_units, output_units,
                 frequency_lower_bound,
                 frequency_upper_bound, approximation_lower_bound,
                 approximation_upper_bound, maximum_error, coefficients,
                 approximation_type='MACLAURIN', resource_id=None,
                 resource_id2=None, name=None,
                 input_units_description=None,
                 output_units_description=None, description=None,
                 decimation_input_sample_rate=None, decimation_factor=None,
                 decimation_offset=None, decimation_delay=None,
                 decimation_correction=None):
        # XXX remove stage_gain and stage_gain_frequency completely, since
        # changes in StationXML 1.1?
        self._approximation_type = approximation_type
        self.frequency_lower_bound = frequency_lower_bound
        self.frequency_upper_bound = frequency_upper_bound
        self.approximation_lower_bound = approximation_lower_bound
        self.approximation_upper_bound = approximation_upper_bound
        self.maximum_error = maximum_error
        self.coefficients = coefficients
        # XXX StationXML 1.1 does not allow stage gain in Polynomial response
        # stages. Maybe we should we warn here.. but this could get very
        # verbose when reading StationXML 1.0 files, so maybe not
        super(PolynomialResponseStage, self).__init__(
            stage_sequence_number=stage_sequence_number,
            input_units=input_units,
            output_units=output_units,
            input_units_description=input_units_description,
            output_units_description=output_units_description,
            resource_id=resource_id, resource_id2=resource_id2,
            stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency, name=name,
            description=description,
            decimation_input_sample_rate=decimation_input_sample_rate,
            decimation_factor=decimation_factor,
            decimation_offset=decimation_offset,
            decimation_delay=decimation_delay,
            decimation_correction=decimation_correction)

    @property
    def approximation_type(self):
        return self._approximation_type

    @approximation_type.setter
    def approximation_type(self, value):
        value = str(value).upper()
        allowed = ("MACLAURIN",)
        if value not in allowed:
            msg = ("Value '%s' for polynomial response approximation type not "
                   "allowed. Possible values are: '%s'")
            msg = msg % (value, "', '".join(allowed))
            raise ValueError(msg)
        self._approximation_type = value

    @property
    def coefficients(self):
        return self._coefficients

    @coefficients.setter
    def coefficients(self, value):
        new_values = []
        for x in value:
            if not isinstance(x, CoefficientWithUncertainties):
                x = CoefficientWithUncertainties(x)
            new_values.append(x)
        self._coefficients = new_values

    def __str__(self):
        ret = super(PolynomialResponseStage, self).__str__()
        ret += (
            "\n"
            "\tPolynomial approximation type: {approximation_type}\n"
            "\tFrequency lower bound: {lower_freq_bound}\n"
            "\tFrequency upper bound: {upper_freq_bound}\n"
            "\tApproximation lower bound: {lower_approx_bound}\n"
            "\tApproximation upper bound: {upper_approx_bound}\n"
            "\tMaximum error: {max_error}\n"
            "\tNumber of coefficients: {coeff_count}".format(
                approximation_type=self._approximation_type,
                lower_freq_bound=self.frequency_lower_bound,
                upper_freq_bound=self.frequency_upper_bound,
                lower_approx_bound=self.approximation_lower_bound,
                upper_approx_bound=self.approximation_upper_bound,
                max_error=self.maximum_error,
                coeff_count=len(self.coefficients)))
        return ret


class Response(ComparingObject):
    """
    The root response object.
    """
    def __init__(self, resource_id=None, instrument_sensitivity=None,
                 instrument_polynomial=None, response_stages=None):
        """
        :type resource_id: str
        :param resource_id: This field contains a string that should serve as a
            unique resource identifier. This identifier can be interpreted
            differently depending on the data center/software that generated
            the document. Also, we recommend to use something like
            GENERATOR:Meaningful ID. As a common behavior equipment with the
            same ID should contains the same information/be derived from the
            same base instruments.
        :type instrument_sensitivity:
            :class:`~obspy.core.inventory.response.InstrumentSensitivity`
        :param instrument_sensitivity: The total sensitivity for the given
            channel, representing the complete acquisition system expressed as
            a scalar.
        :type instrument_polynomial:
            :class:`~obspy.core.inventory.response.InstrumentPolynomial`
        :param instrument_polynomial: The total sensitivity for the given
            channel, representing the complete acquisition system expressed as
            a polynomial.
        :type response_stages: list of
            :class:`~obspy.core.inventory.response.ResponseStage` objects
        :param response_stages: A list of the response stages. Covers SEED
            blockettes 53 to 56.
        """
        self.resource_id = resource_id
        self.instrument_sensitivity = instrument_sensitivity
        self.instrument_polynomial = instrument_polynomial
        if response_stages is None:
            self.response_stages = []
        elif hasattr(response_stages, "__iter__"):
            self.response_stages = response_stages
        else:
            msg = "response_stages must be an iterable."
            raise ValueError(msg)

    def _attempt_to_fix_units(self):
        """
        Internal helper function that will add units to gain only stages based
        on the units of surrounding stages.

        Should be called when parsing from file formats that don't have units
        for identity stages.
        """
        previous_output_units = None
        previous_output_units_description = None

        # Potentially set the input units of the first stage to the units of
        # the overall sensitivity and the output units of the second stage.
        if self.response_stages and self.response_stages[0] and \
                hasattr(self, "instrument_sensitivity"):
            s = self.instrument_sensitivity

            if s:
                if self.response_stages[0].input_units is None:
                    self.response_stages[0].input_units = s.input_units
                if self.response_stages[0].input_units_description is None:
                    self.response_stages[0].input_units_description = \
                        s.input_units_description

            if len(self.response_stages) >= 2 and self.response_stages[1]:
                if self.response_stages[0].output_units is None:
                    self.response_stages[0].output_units = \
                        self.response_stages[1].input_units
                if self.response_stages[0].output_units_description is None:
                    self.response_stages[0].output_units_description = \
                        self.response_stages[1].input_units_description

        # Front to back.
        for r in self.response_stages:
            # Only for identity/stage only.
            if type(r) is ResponseStage:
                if not r.input_units and not r.output_units and \
                        previous_output_units:
                    r.input_units = previous_output_units
                    r.output_units = previous_output_units
                if not r.input_units_description and \
                        not r.output_units_description \
                        and previous_output_units_description:
                    r.input_units_description = \
                        previous_output_units_description
                    r.output_units_description = \
                        previous_output_units_description

            previous_output_units = r.output_units
            previous_output_units_description = r.output_units_description

        # Back to front.
        previous_input_units = None
        previous_input_units_description = None
        for r in reversed(self.response_stages):
            # Only for identity/stage only.
            if type(r) is ResponseStage:
                if not r.input_units and not r.output_units and \
                        previous_input_units:
                    r.input_units = previous_input_units
                    r.output_units = previous_input_units
                if not r.input_units_description and \
                        not r.output_units_description \
                        and previous_input_units_description:
                    r.input_units_description = \
                        previous_input_units_description
                    r.output_units_description = \
                        previous_input_units_description

            previous_input_units = r.input_units
            previous_input_units_description = r.input_units_description

    def get_sampling_rates(self):
        """
        Computes the input and output sampling rates of each stage.

        For well defined files this will just read the decimation attributes
        of each stage. For others it will attempt to infer missing values
        from the surrounding stages.

        :returns: A nested dictionary detailing the sampling rates of each
            response stage.
        :rtype: dict

        >>> import obspy
        >>> inv = obspy.read_inventory("AU.MEEK.xml")  # doctest: +SKIP
        >>> inv[0][0][0].response.get_sampling_rates()  # doctest: +SKIP
        {1: {'decimation_factor': 1,
             'input_sampling_rate': 600.0,
             'output_sampling_rate': 600.0},
         2: {'decimation_factor': 1,
             'input_sampling_rate': 600.0,
             'output_sampling_rate': 600.0},
         3: {'decimation_factor': 1,
             'input_sampling_rate': 600.0,
             'output_sampling_rate': 600.0},
         4: {'decimation_factor': 3,
             'input_sampling_rate': 600.0,
             'output_sampling_rate': 200.0},
         5: {'decimation_factor': 10,
             'input_sampling_rate': 200.0,
             'output_sampling_rate': 20.0}}
        """
        # Get all stages, but skip stage 0.
        stages = [_i.stage_sequence_number for _i in self.response_stages
                  if _i.stage_sequence_number]
        if not stages:
            return {}

        if list(range(1, len(stages) + 1)) != stages:
            raise ValueError("Can only determine sampling rates if response "
                             "stages are in order.")

        # First fill in all the set values.
        sampling_rates = {}
        for stage in self.response_stages:
            input_sr = None
            output_sr = None
            factor = None
            if stage.decimation_input_sample_rate:
                input_sr = stage.decimation_input_sample_rate
                if stage.decimation_factor:
                    factor = stage.decimation_factor
                    output_sr = input_sr / float(factor)
            sampling_rates[stage.stage_sequence_number] = {
                "input_sampling_rate": input_sr,
                "output_sampling_rate": output_sr,
                "decimation_factor": factor}

        # Nothing might be set - just return in that case.
        if set(itertools.chain.from_iterable(v.values()
               for v in sampling_rates.values())) == {None}:
            return sampling_rates

        # Find the first set input sampling rate. The output sampling rate
        # cannot be set without it. Set all prior input and output sampling
        # rates to it.
        for i in stages:
            sr = sampling_rates[i]["input_sampling_rate"]
            if sr:
                for j in range(1, i):
                    sampling_rates[j]["input_sampling_rate"] = sr
                    sampling_rates[j]["output_sampling_rate"] = sr
                    sampling_rates[j]["decimation_factor"] = 1
                break

        # This should guarantee that the input and output sampling rate of the
        # the first stage are set.
        output_sr = sampling_rates[1]["output_sampling_rate"]
        if not output_sr:  # pragma: no cover
            raise NotImplementedError

        for i in stages:
            si = sampling_rates[i]
            if not si["input_sampling_rate"]:
                si["input_sampling_rate"] = output_sr
            if not si["output_sampling_rate"]:
                if not si["decimation_factor"]:
                    si["output_sampling_rate"] = si["input_sampling_rate"]
                    si["decimation_factor"] = 1
                else:
                    si["output_sampling_rate"] = si["input_sampling_rate"] / \
                        float(si["decimation_factor"])
            if not si["decimation_factor"]:
                si["decimation_factor"] = int(round(
                    si["input_sampling_rate"] / si["output_sampling_rate"]))
            output_sr = si["output_sampling_rate"]

        def is_close(a, b):
            return abs(a - b) < 1e-5

        # Final consistency checks.
        sr = sampling_rates[stages[0]]["input_sampling_rate"]
        for i in stages:
            si = sampling_rates[i]
            if not is_close(si["input_sampling_rate"], sr):  # pragma: no cover
                msg = ("Input sampling rate of stage %i is inconsistent "
                       "with the previous stages' output sampling rate")
                warnings.warn(msg % i)

            if not is_close(
                    si["input_sampling_rate"] / si["output_sampling_rate"],
                    si["decimation_factor"]):  # pragma: no cover
                msg = ("Decimation factor in stage %i is inconsistent with "
                       "input and output sampling rates.")
                warnings.warn(msg % i)
            sr = si["output_sampling_rate"]

        return sampling_rates

    def recalculate_overall_sensitivity(self, frequency=None):
        """
        Recalculates the overall sensitivity.

        :param frequency: Choose frequency at which to calculate the
            sensitivity. If not given it will be chosen automatically.
        """
        if not hasattr(self, "instrument_sensitivity"):
            msg = "Could not find an instrument sensitivity - will not " \
                  "recalculate the overall sensitivity."
            raise ValueError(msg)

        if not self.instrument_sensitivity.input_units:
            msg = "Could not determine input units - will not " \
                  "recalculate the overall sensitivity."
            raise ValueError(msg)

        i_u = self.instrument_sensitivity.input_units

        unit_map = {
            "DISP": ["M"],
            "VEL": ["M/S", "M/SEC"],
            "ACC": ["M/S**2", "M/(S**2)", "M/SEC**2", "M/(SEC**2)",
                    "M/S/S"]}
        unit = None
        for key, value in unit_map.items():
            if i_u and i_u.upper() in value:
                unit = key
        if not unit:
            msg = ("ObsPy does not know how to map unit '%s' to "
                   "displacement, velocity, or acceleration - overall "
                   "sensitivity will not be recalculated.") % i_u
            raise ValueError(msg)

        # Determine frequency if not given.
        if frequency is None:
            # lookup normalization frequency of sensor's first stage it should
            # be in the flat part of the response
            stage_one = self.response_stages[0]
            try:
                frequency = stage_one.normalization_frequency
            except AttributeError:
                pass
            for stage in self.response_stages[::-1]:
                # determine sampling rate
                try:
                    sampling_rate = (stage.decimation_input_sample_rate /
                                     stage.decimation_factor)
                    break
                except Exception:
                    continue
            else:
                sampling_rate = None
            if sampling_rate:
                # if sensor's normalization frequency is above 0.5 * nyquist,
                # use that instead (e.g. to avoid computing an overall
                # sensitivity above nyquist)
                nyquist = sampling_rate / 2.0
                if frequency:
                    frequency = min(frequency, nyquist / 2.0)
                else:
                    frequency = nyquist / 2.0

        if frequency is None:
            msg = ("Could not automatically determine a suitable frequency "
                   "at which to calculate the sensitivity. The overall "
                   "sensitivity will not be recalculated.")
            raise ValueError(msg)

        freq, gain = self._get_overall_sensitivity_and_gain(
            output=unit, frequency=float(frequency))

        self.instrument_sensitivity.value = gain
        self.instrument_sensitivity.frequency = freq

    def _get_overall_sensitivity_and_gain(
            self, frequency=None, output='VEL'):
        """
        Get the overall sensitivity and gain from stages 1 to N.

        Returns the overall sensitivity frequency and gain, which can be
        used to create stage 0.

        :type output: str
        :param output: Output units. One of:

            ``"DISP"``
                displacement, output unit is meters
            ``"VEL"``
                velocity, output unit is meters/second
            ``"ACC"``
                acceleration, output unit is meters/second**2
            ``"DEF"``
                default units, the response is calculated in
                output units/input units (last stage/first stage).
                Useful if the units for a particular type of sensor (e.g., a
                pressure sensor) cannot be converted to displacement, velocity
                or acceleration.

        :type frequency: float
        :param frequency: Frequency to calculate overall sensitivity for in
            Hertz. Defaults to normalization frequency of stage 1.
        :rtype: :tuple: ( float, float )
        :returns: frequency and gain at frequency.
        """
        if frequency is None:
            # XXX is this safe enough, or should we lookup the stage sequence
            # XXX number explicitly?
            frequency = self.response_stages[0].normalization_frequency
        response_at_frequency = self._call_eval_resp_for_frequencies(
            frequencies=[frequency], output=output,
            hide_sensitivity_mismatch_warning=True)[0][0]
        overall_sensitivity = abs(response_at_frequency)
        return frequency, overall_sensitivity

    def _call_eval_resp_for_frequencies(
            self, frequencies, output="VEL", start_stage=None,
            end_stage=None, hide_sensitivity_mismatch_warning=False):
        """
        Returns frequency response for given frequencies using evalresp.

        Also returns the overall sensitivity frequency and its gain.

        :type frequencies: list[float]
        :param frequencies: Discrete frequencies to calculate response for.
        :type output: str
        :param output: Output units. One of:

            ``"DISP"``
                displacement, output unit is meters
            ``"VEL"``
                velocity, output unit is meters/second
            ``"ACC"``
                acceleration, output unit is meters/second**2
            ``"DEF"``
                default units, the response is calculated in
                output units/input units (last stage/first stage).
                Useful if the units for a particular type of sensor (e.g., a
                pressure sensor) cannot be converted to displacement, velocity
                or acceleration.

        :type start_stage: int, optional
        :param start_stage: Stage sequence number of first stage that will be
            used (disregarding all earlier stages).
        :type end_stage: int, optional
        :param end_stage: Stage sequence number of last stage that will be
            used (disregarding all later stages).
        :type hide_sensitivity_mismatch_warning: bool
        :param hide_sensitivity_mismatch_warning: Hide the evalresp warning
            that computed and reported sensitivities don't match.
        :rtype: tuple(:class:`numpy.ndarray`, chan)
        :returns: frequency response at requested frequencies
        """
        if not self.response_stages:
            msg = ("Can not use evalresp on response with no response "
                   "stages.")
            raise ObsPyException(msg)

        import scipy.interpolate
        import obspy.signal.evrespwrapper as ew
        from obspy.signal.headers import clibevresp

        out_units = output.upper()
        if out_units not in ("DISP", "VEL", "ACC", "DEF"):
            msg = ("requested output is '%s' but must be one of 'DISP', 'VEL' "
                   ", 'ACC' or 'DEF'") % output
            raise ValueError(msg)

        frequencies = np.asarray(frequencies)

        # Whacky. Evalresp uses a global variable and uses that to scale the
        # response if it encounters any unit that is not SI.
        scale_factor = [1.0]

        def get_unit_mapping(key):
            try:
                key = key.upper()
            except Exception:
                pass
            units_mapping = {
                "M": ew.ENUM_UNITS["DIS"],
                "NM": ew.ENUM_UNITS["DIS"],
                "CM": ew.ENUM_UNITS["DIS"],
                "MM": ew.ENUM_UNITS["DIS"],
                "M/S": ew.ENUM_UNITS["VEL"],
                "M/SEC": ew.ENUM_UNITS["VEL"],
                "NM/S": ew.ENUM_UNITS["VEL"],
                "NM/SEC": ew.ENUM_UNITS["VEL"],
                "CM/S": ew.ENUM_UNITS["VEL"],
                "CM/SEC": ew.ENUM_UNITS["VEL"],
                "MM/S": ew.ENUM_UNITS["VEL"],
                "MM/SEC": ew.ENUM_UNITS["VEL"],
                "M/S**2": ew.ENUM_UNITS["ACC"],
                "M/(S**2)": ew.ENUM_UNITS["ACC"],
                "M/SEC**2": ew.ENUM_UNITS["ACC"],
                "M/(SEC**2)": ew.ENUM_UNITS["ACC"],
                "M/S/S": ew.ENUM_UNITS["ACC"],
                "NM/S**2": ew.ENUM_UNITS["ACC"],
                "NM/(S**2)": ew.ENUM_UNITS["ACC"],
                "NM/SEC**2": ew.ENUM_UNITS["ACC"],
                "NM/(SEC**2)": ew.ENUM_UNITS["ACC"],
                "CM/S**2": ew.ENUM_UNITS["ACC"],
                "CM/(S**2)": ew.ENUM_UNITS["ACC"],
                "CM/SEC**2": ew.ENUM_UNITS["ACC"],
                "CM/(SEC**2)": ew.ENUM_UNITS["ACC"],
                "MM/S**2": ew.ENUM_UNITS["ACC"],
                "MM/(S**2)": ew.ENUM_UNITS["ACC"],
                "MM/SEC**2": ew.ENUM_UNITS["ACC"],
                "MM/(SEC**2)": ew.ENUM_UNITS["ACC"],
                # Evalresp internally treats strain as displacement.
                "M/M": ew.ENUM_UNITS["DIS"],
                "M**3/M**3": ew.ENUM_UNITS["DIS"],
                "V": ew.ENUM_UNITS["VOLTS"],
                "VOLT": ew.ENUM_UNITS["VOLTS"],
                "VOLTS": ew.ENUM_UNITS["VOLTS"],
                # This is weird, but evalresp appears to do the same.
                "V/M": ew.ENUM_UNITS["VOLTS"],
                "COUNT": ew.ENUM_UNITS["COUNTS"],
                "COUNTS": ew.ENUM_UNITS["COUNTS"],
                "T": ew.ENUM_UNITS["TESLA"],
                "PA": ew.ENUM_UNITS["PRESSURE"],
                "PASCAL": ew.ENUM_UNITS["PRESSURE"],
                "PASCALS": ew.ENUM_UNITS["PRESSURE"],
                "MBAR": ew.ENUM_UNITS["PRESSURE"]}
            if key not in units_mapping:
                if key is not None:
                    msg = ("The unit '%s' is not known to ObsPy. It will be "
                           "assumed to be displacement for the calculations. "
                           "This mostly does the right thing but please "
                           "proceed with caution.") % key
                    warnings.warn(msg)
                value = ew.ENUM_UNITS["DIS"]
            else:
                value = units_mapping[key]

            # Scale factor with the same logic as evalresp.
            if key in ["CM/S**2", "CM/S", "CM/SEC", "CM"]:
                scale_factor[0] = 1.0E2
            elif key in ["MM/S**2", "MM/S", "MM/SEC", "MM"]:
                scale_factor[0] = 1.0E3
            elif key in ["NM/S**2", "NM/S", "NM/SEC", "NM"]:
                scale_factor[0] = 1.0E9

            return value

        all_stages = defaultdict(list)

        for stage in self.response_stages:
            # optionally select only stages as requested by user
            if start_stage is not None:
                if stage.stage_sequence_number < start_stage:
                    continue
            if end_stage is not None:
                if stage.stage_sequence_number > end_stage:
                    continue
            all_stages[stage.stage_sequence_number].append(stage)

        stage_lengths = set(map(len, all_stages.values()))
        if len(stage_lengths) != 1 or stage_lengths.pop() != 1:
            msg = "Each stage can only appear once."
            raise ValueError(msg)

        stage_list = sorted(all_stages.keys())

        stage_objects = []

        # Attempt to fix some potentially faulty responses here.
        if 1 in all_stages and all_stages[1] and (
                not all_stages[1][0].input_units or
                not all_stages[1][0].output_units):
            # Make a copy to not modify the original
            all_stages[1][0] = copy.deepcopy(all_stages[1][0])
            # Some stages 1 are just the sensitivity and as thus don't store
            # input and output units in for example StationXML. In these cases
            # try to guess it from the overall sensitivity or stage 2.
            if not all_stages[1][0].input_units:
                if self.instrument_sensitivity.input_units:
                    all_stages[1][0].input_units = \
                        self.instrument_sensitivity.input_units
                    msg = "Set the input units of stage 1 to the overall " \
                        "input units."
                    warnings.warn(msg)
            if not all_stages[1][0].output_units:
                if max(all_stages.keys()) == 1 and \
                        self.instrument_sensitivity.output_units:
                    all_stages[1][0].output_units = \
                        self.instrument_sensitivity.output_units
                    msg = "Set the output units of stage 1 to the overall " \
                        "output units."
                    warnings.warn(msg)
                if 2 in all_stages and all_stages[2] and \
                        all_stages[2][0].input_units:
                    all_stages[1][0].output_units = \
                        all_stages[2][0].input_units
                    msg = "Set the output units of stage 1 to the input " \
                        "units of stage 2."
                    warnings.warn(msg)

        for stage_number in stage_list:
            st = ew.Stage()
            st.sequence_no = stage_number

            stage_blkts = []

            blockette = all_stages[stage_number][0]

            # Write the input and output units.
            st.input_units = get_unit_mapping(blockette.input_units)
            st.output_units = get_unit_mapping(blockette.output_units)

            if isinstance(blockette, PolesZerosResponseStage):
                blkt = ew.Blkt()
                # Map the transfer function type.
                transfer_fct_mapping = {
                    "LAPLACE (RADIANS/SECOND)": "LAPLACE_PZ",
                    "LAPLACE (HERTZ)": "ANALOG_PZ",
                    "DIGITAL (Z-TRANSFORM)": "IIR_PZ"}
                blkt.type = ew.ENUM_FILT_TYPES[transfer_fct_mapping[
                    blockette.pz_transfer_function_type]]

                # The blockette is a pole zero blockette.
                pz = blkt.blkt_info.pole_zero

                pz.nzeros = len(blockette.zeros)
                pz.npoles = len(blockette.poles)
                pz.a0 = blockette.normalization_factor
                pz.a0_freq = blockette.normalization_frequency

                # XXX: Find a better way to do this.
                poles = (ew.ComplexNumber * len(blockette.poles))()
                for i, value in enumerate(blockette.poles):
                    poles[i].real = value.real
                    poles[i].imag = value.imag

                zeros = (ew.ComplexNumber * len(blockette.zeros))()
                for i, value in enumerate(blockette.zeros):
                    zeros[i].real = value.real
                    zeros[i].imag = value.imag

                pz.poles = C.cast(C.pointer(poles),
                                  C.POINTER(ew.ComplexNumber))
                pz.zeros = C.cast(C.pointer(zeros),
                                  C.POINTER(ew.ComplexNumber))
            elif isinstance(blockette, CoefficientsTypeResponseStage):
                blkt = ew.Blkt()
                # This type can have either an FIR or an IIR response. If
                # the number of denominators is 0, it is a FIR. Otherwise
                # an IIR.

                # FIR
                if len(blockette.denominator) == 0:
                    if blockette.cf_transfer_function_type.lower() \
                            != "digital":
                        msg = ("When no denominators are given it must "
                               "be a digital FIR filter.")
                        raise ValueError(msg)
                    # Set the type to an asymmetric FIR blockette.
                    blkt.type = ew.ENUM_FILT_TYPES["FIR_ASYM"]
                    fir = blkt.blkt_info.fir
                    fir.h0 = 1.0
                    fir.ncoeffs = len(blockette.numerator)
                    # XXX: Find a better way to do this.
                    coeffs = (C.c_double * len(blockette.numerator))()
                    for i, value in enumerate(blockette.numerator):
                        coeffs[i] = float(value)
                    fir.coeffs = C.cast(C.pointer(coeffs),
                                        C.POINTER(C.c_double))
                # IIR
                else:
                    blkt.type = ew.ENUM_FILT_TYPES["IIR_COEFFS"]
                    coeff = blkt.blkt_info.coeff

                    coeff.h0 = 1.0
                    coeff.nnumer = len(blockette.numerator)
                    coeff.ndenom = len(blockette.denominator)

                    # XXX: Find a better way to do this.
                    coeffs = (C.c_double * len(blockette.numerator))()
                    for i, value in enumerate(blockette.numerator):
                        coeffs[i] = float(value)
                    coeff.numer = C.cast(C.pointer(coeffs),
                                         C.POINTER(C.c_double))
                    coeffs = (C.c_double * len(blockette.denominator))()
                    for i, value in enumerate(blockette.denominator):
                        coeffs[i] = float(value)
                    coeff.denom = C.cast(C.pointer(coeffs),
                                         C.POINTER(C.c_double))
            elif isinstance(blockette, ResponseListResponseStage):
                blkt = ew.Blkt()
                blkt.type = ew.ENUM_FILT_TYPES["LIST"]

                # Get values as numpy arrays.
                f = np.array([float(_i.frequency)
                              for _i in blockette.response_list_elements],
                             dtype=np.float64)
                amp = np.array([float(_i.amplitude)
                                for _i in blockette.response_list_elements],
                               dtype=np.float64)
                phase = np.array([
                    float(_i.phase)
                    for _i in blockette.response_list_elements],
                    dtype=np.float64)

                # Sanity check.
                min_f = frequencies[frequencies > 0].min()
                max_f = frequencies.max()

                min_f_avail = min(f)
                max_f_avail = max(f)

                if min_f < min_f_avail or max_f > max_f_avail:
                    msg = (
                        "The response contains a "
                        "response list stage with frequencies only from "
                        "%.4f - %.4f Hz. You are requesting a response from "
                        "%.4f - %.4f Hz. The calculated response will contain "
                        "extrapolation beyond the frequency band constrained "
                        "by the response list stage. Please consider "
                        "adjusting 'pre_filt' and/or 'water_level' during "
                        "response removal accordingly.")
                    warnings.warn(
                        msg % (min_f_avail, max_f_avail, min_f, max_f))

                amp = scipy.interpolate.InterpolatedUnivariateSpline(
                    f, amp, k=3)(frequencies)
                phase = scipy.interpolate.InterpolatedUnivariateSpline(
                    f, phase, k=3)(frequencies)

                # Set static offset to zero.
                amp[amp == 0] = 0
                phase[phase == 0] = 0

                rl = blkt.blkt_info.list
                rl.nresp = len(frequencies)

                _freq_c = (C.c_double * rl.nresp)()
                _amp_c = (C.c_double * rl.nresp)()
                _phase_c = (C.c_double * rl.nresp)()

                _freq_c[:] = frequencies[:]
                _amp_c[:] = amp[:]
                _phase_c[:] = phase[:]

                rl.freq = C.cast(C.pointer(_freq_c),
                                 C.POINTER(C.c_double))
                rl.amp = C.cast(C.pointer(_amp_c),
                                C.POINTER(C.c_double))
                rl.phase = C.cast(C.pointer(_phase_c),
                                  C.POINTER(C.c_double))

            elif isinstance(blockette, FIRResponseStage):
                blkt = ew.Blkt()

                if blockette.symmetry == "NONE":
                    blkt.type = ew.ENUM_FILT_TYPES["FIR_ASYM"]
                if blockette.symmetry == "ODD":
                    blkt.type = ew.ENUM_FILT_TYPES["FIR_SYM_1"]
                if blockette.symmetry == "EVEN":
                    blkt.type = ew.ENUM_FILT_TYPES["FIR_SYM_2"]

                # The blockette is a fir blockette
                fir = blkt.blkt_info.fir
                fir.h0 = 1.0
                fir.ncoeffs = len(blockette.coefficients)

                # XXX: Find a better way to do this.
                coeffs = (C.c_double * len(blockette.coefficients))()
                for i, value in enumerate(blockette.coefficients):
                    coeffs[i] = float(value)
                fir.coeffs = C.cast(C.pointer(coeffs),
                                    C.POINTER(C.c_double))
            elif isinstance(blockette, PolynomialResponseStage):
                msg = ("PolynomialResponseStage not yet implemented. "
                       "Please contact the developers.")
                raise NotImplementedError(msg)
            else:
                # Otherwise it could be a gain only stage.
                if blockette.stage_gain is not None and \
                        blockette.stage_gain_frequency is not None:
                    blkt = None
                else:
                    msg = "Type: %s." % str(type(blockette))
                    raise NotImplementedError(msg)

            if blkt is not None:
                stage_blkts.append(blkt)

            # Evalresp requires FIR and IIR blockettes to have decimation
            # values. Set the "unit decimation" values in case they are not
            # set.
            #
            # Only set it if there is a stage gain - otherwise evalresp
            # complains again.
            if isinstance(blockette, PolesZerosResponseStage) and \
                    blockette.stage_gain and \
                    None in set([
                        blockette.decimation_correction,
                        blockette.decimation_delay,
                        blockette.decimation_factor,
                        blockette.decimation_input_sample_rate,
                        blockette.decimation_offset]):
                # Don't modify the original object.
                blockette = copy.deepcopy(blockette)
                blockette.decimation_correction = 0.0
                blockette.decimation_delay = 0.0
                blockette.decimation_factor = 1
                blockette.decimation_offset = 0
                sr = self.get_sampling_rates()
                if sr and blockette.stage_sequence_number in sr and \
                        sr[blockette.stage_sequence_number][
                            "input_sampling_rate"]:
                    blockette.decimation_input_sample_rate = \
                        self.get_sampling_rates()[
                            blockette.stage_sequence_number][
                            "input_sampling_rate"]
                # This branch get's large called for responses that only have a
                # a single stage.
                else:
                    blockette.decimation_input_sample_rate = 1.0

            # Parse the decimation if is given.
            decimation_values = set([
                blockette.decimation_correction,
                blockette.decimation_delay, blockette.decimation_factor,
                blockette.decimation_input_sample_rate,
                blockette.decimation_offset])
            if None in decimation_values:
                if len(decimation_values) != 1:
                    msg = ("If a decimation is given, all values must "
                           "be specified.")
                    raise ValueError(msg)
            else:
                blkt = ew.Blkt()
                blkt.type = ew.ENUM_FILT_TYPES["DECIMATION"]
                decimation_blkt = blkt.blkt_info.decimation

                # Evalresp does the same!
                if blockette.decimation_input_sample_rate == 0:
                    decimation_blkt.sample_int = 0.0
                else:
                    decimation_blkt.sample_int = \
                        1.0 / blockette.decimation_input_sample_rate

                decimation_blkt.deci_fact = blockette.decimation_factor
                decimation_blkt.deci_offset = blockette.decimation_offset
                decimation_blkt.estim_delay = blockette.decimation_delay
                decimation_blkt.applied_corr = \
                    blockette.decimation_correction
                stage_blkts.append(blkt)

            # Add the gain if it is available.
            if blockette.stage_gain is not None and \
                    blockette.stage_gain_frequency is not None:
                blkt = ew.Blkt()
                blkt.type = ew.ENUM_FILT_TYPES["GAIN"]
                gain_blkt = blkt.blkt_info.gain
                gain_blkt.gain = blockette.stage_gain
                gain_blkt.gain_freq = blockette.stage_gain_frequency
                stage_blkts.append(blkt)

            if not stage_blkts:
                msg = "At least one blockette is needed for the stage."
                raise ValueError(msg)

            # Attach the blockette chain to the stage.
            st.first_blkt = C.pointer(stage_blkts[0])
            for _i in range(1, len(stage_blkts)):
                stage_blkts[_i - 1].next_blkt = C.pointer(stage_blkts[_i])

            stage_objects.append(st)

        # Attach the instrument sensitivity as stage 0 at the end.
        st = ew.Stage()
        st.sequence_no = 0
        st.input_units = 0
        st.output_units = 0
        blkt = ew.Blkt()
        blkt.type = ew.ENUM_FILT_TYPES["GAIN"]
        gain_blkt = blkt.blkt_info.gain
        gain_blkt.gain = self.instrument_sensitivity.value
        # Set the sensitivity frequency - use 1.0 if not given. This is also
        # what evalresp does.
        gain_blkt.gain_freq = self.instrument_sensitivity.frequency \
            if self.instrument_sensitivity.frequency else 0.0
        st.first_blkt = C.pointer(blkt)
        stage_objects.append(st)

        chan = ew.Channel()
        if not stage_objects:
            msg = "At least one stage is needed."
            raise ValueError(msg)

        # Attach the stage chain to the channel.
        chan.first_stage = C.pointer(stage_objects[0])
        for _i in range(1, len(stage_objects)):
            stage_objects[_i - 1].next_stage = C.pointer(stage_objects[_i])

        chan.nstages = len(stage_objects)

        # Evalresp will take care of setting it to the overall sensitivity.
        chan.sensit = 0.0
        chan.sensfreq = 0.0

        output = np.empty(len(frequencies), dtype=np.complex128)
        out_units = C.c_char_p(out_units.encode('ascii', 'strict'))

        # Set global variables
        if self.resource_id:
            clibevresp.curr_file.value = self.resource_id.encode('utf-8')
        else:
            clibevresp.curr_file.value = None

        try:
            rc = clibevresp._obspy_check_channel(C.byref(chan))
            if rc:
                e, m = ew.ENUM_ERROR_CODES[rc]
                raise e('check_channel: ' + m)
            rc = clibevresp._obspy_norm_resp(
                C.byref(chan), -1, 0,
                1 if hide_sensitivity_mismatch_warning else 0)
            if rc:
                e, m = ew.ENUM_ERROR_CODES[rc]
                raise e('norm_resp: ' + m)

            rc = clibevresp._obspy_calc_resp(C.byref(chan), frequencies,
                                             len(frequencies),
                                             output, out_units, -1, 0, 0)
            if rc:
                e, m = ew.ENUM_ERROR_CODES[rc]
                raise e('calc_resp: ' + m)

            # XXX: Check if this is really not needed.
            # output *= scale_factor[0]

        finally:
            clibevresp.curr_file.value = None

        return output, chan

    def get_evalresp_response_for_frequencies(
            self, frequencies, output="VEL", start_stage=None, end_stage=None,
            hide_sensitivity_mismatch_warning=False):
        """
        Returns frequency response for given frequencies using evalresp.

        :type frequencies: list[float]
        :param frequencies: Discrete frequencies to calculate response for.
        :type output: str
        :param output: Output units. One of:

            ``"DISP"``
                displacement, output unit is meters
            ``"VEL"``
                velocity, output unit is meters/second
            ``"ACC"``
                acceleration, output unit is meters/second**2
            ``"DEF"``
                default units, the response is calculated in
                output units/input units (last stage/first stage).
                Useful if the units for a particular type of sensor (e.g., a
                pressure sensor) cannot be converted to displacement, velocity
                or acceleration.

        :type start_stage: int, optional
        :param start_stage: Stage sequence number of first stage that will be
            used (disregarding all earlier stages).
        :type end_stage: int, optional
        :param end_stage: Stage sequence number of last stage that will be
            used (disregarding all later stages).
        :type hide_sensitivity_mismatch_warning: bool
        :param hide_sensitivity_mismatch_warning: Hide the evalresp warning
            that computed and reported sensitivities do not match.
        :rtype: :class:`numpy.ndarray`
        :returns: frequency response at requested frequencies
        """
        hsmw = hide_sensitivity_mismatch_warning  # PEP8
        output, chan = self._call_eval_resp_for_frequencies(
            frequencies, output=output, start_stage=start_stage,
            end_stage=end_stage,
            hide_sensitivity_mismatch_warning=hsmw)
        return output

    def get_evalresp_response(self, t_samp, nfft, output="VEL",
                              start_stage=None, end_stage=None,
                              hide_sensitivity_mismatch_warning=False):
        """
        Returns frequency response and corresponding frequencies using
        evalresp.

        :type t_samp: float
        :param t_samp: time resolution (inverse frequency resolution)
        :type nfft: int
        :param nfft: Number of FFT points to use
        :type output: str
        :param output: Output units. One of:

            ``"DISP"``
                displacement, output unit is meters
            ``"VEL"``
                velocity, output unit is meters/second
            ``"ACC"``
                acceleration, output unit is meters/second**2
            ``"DEF"``
                default units, the response is calculated in
                output units/input units (last stage/first stage).
                Useful if the units for a particular type of sensor (e.g., a
                pressure sensor) cannot be converted to displacement, velocity
                or acceleration.

        :type start_stage: int, optional
        :param start_stage: Stage sequence number of first stage that will be
            used (disregarding all earlier stages).
        :type end_stage: int, optional
        :param end_stage: Stage sequence number of last stage that will be
            used (disregarding all later stages).
        :type hide_sensitivity_mismatch_warning: bool
        :param hide_sensitivity_mismatch_warning: Hide the evalresp warning
            that computed and reported sensitivities do not match.
        :rtype: tuple(:class:`numpy.ndarray`, :class:`numpy.ndarray`)
        :returns: frequency response and corresponding frequencies
        """
        # Calculate the output frequencies.
        fy = 1 / (t_samp * 2.0)
        # start at zero to get zero for offset/ DC of fft
        # numpy 1.9 introduced a dtype kwarg
        try:
            freqs = np.linspace(0, fy, int(nfft // 2) + 1, dtype=np.float64)
        except Exception:
            freqs = np.linspace(0, fy, int(nfft // 2) + 1).astype(np.float64)

        hsmw = hide_sensitivity_mismatch_warning  # PEP8
        response = self.get_evalresp_response_for_frequencies(
            freqs, output=output, start_stage=start_stage, end_stage=end_stage,
            hide_sensitivity_mismatch_warning=hsmw)
        return response, freqs

    def __str__(self):
        i_s = self.instrument_sensitivity
        if i_s:
            input_units = i_s.input_units \
                if i_s.input_units else "UNKNOWN"
            input_units_description = i_s.input_units_description \
                if i_s.input_units_description else ""
            output_units = i_s.output_units \
                if i_s.output_units else "UNKNOWN"
            output_units_description = i_s.output_units_description \
                if i_s.output_units_description else ""
            sensitivity = ("%g" % i_s.value) if i_s.value else "UNKNOWN"
            freq = ("%.3f" % i_s.frequency) if i_s.frequency else "UNKNOWN"
        else:
            input_units = "UNKNOWN"
            input_units_description = ""
            output_units = "UNKNOWN"
            output_units_description = ""
            sensitivity = "UNKNOWN"
            freq = "UNKNOWN"

        ret = (
            "Channel Response\n"
            "\tFrom {input_units} ({input_units_description}) to "
            "{output_units} ({output_units_description})\n"
            "\tOverall Sensitivity: {sensitivity} defined at {freq} Hz\n"
            "\t{stages} stages:\n{stage_desc}").format(
            input_units=input_units,
            input_units_description=input_units_description,
            output_units=output_units,
            output_units_description=output_units_description,
            sensitivity=sensitivity,
            freq=freq,
            stages=len(self.response_stages),
            stage_desc="\n".join(
                ["\t\tStage %i: %s from %s to %s,"
                 " gain: %s" % (
                     i.stage_sequence_number, i.__class__.__name__,
                     i.input_units, i.output_units,
                     ("%g" % i.stage_gain) if i.stage_gain else "UNKNOWN")
                 for i in self.response_stages]))
        return ret

    def _repr_pretty_(self, p, cycle):
        p.text(str(self))

    def plot(self, min_freq, output="VEL", start_stage=None,
             end_stage=None, label=None, axes=None, sampling_rate=None,
             unwrap_phase=False, plot_degrees=False, show=True, outfile=None):
        """
        Show bode plot of instrument response.

        :type min_freq: float
        :param min_freq: Lowest frequency to plot.
        :type output: str
        :param output: Output units. One of:

            ``"DISP"``
                displacement, output unit is meters
            ``"VEL"``
                velocity, output unit is meters/second
            ``"ACC"``
                acceleration, output unit is meters/second**2
            ``"DEF"``
                default units, the response is calculated in
                output units/input units (last stage/first stage).
                Useful if the units for a particular type of sensor (e.g., a
                pressure sensor) cannot be converted to displacement, velocity
                or acceleration.

        :type start_stage: int, optional
        :param start_stage: Stage sequence number of first stage that will be
            used (disregarding all earlier stages).
        :type end_stage: int, optional
        :param end_stage: Stage sequence number of last stage that will be
            used (disregarding all later stages).
        :type label: str
        :param label: Label string for legend.
        :type axes: list of 2 :class:`matplotlib.axes.Axes`
        :param axes: List/tuple of two axes instances to plot the
            amplitude/phase spectrum into. If not specified, a new figure is
            opened.
        :type sampling_rate: float
        :param sampling_rate: Manually specify sampling rate of time series.
            If not given it is attempted to determine it from the information
            in the individual response stages.  Does not influence the spectra
            calculation, if it is not known, just provide the highest frequency
            that should be plotted times two.
        :type unwrap_phase: bool
        :param unwrap_phase: Set optional phase unwrapping using NumPy.
        :type plot_degrees: bool
        :param plot_degrees: if ``True`` plot bode in degrees
        :type show: bool
        :param show: Whether to show the figure after plotting or not. Can be
            used to do further customization of the plot before showing it.
        :type outfile: str
        :param outfile: Output file path to directly save the resulting image
            (e.g. ``"/tmp/image.png"``). Overrides the ``show`` option, image
            will not be displayed interactively. The given path/filename is
            also used to automatically determine the output format. Supported
            file formats depend on your matplotlib backend.  Most backends
            support png, pdf, ps, eps and svg. Defaults to ``None``.

        .. rubric:: Basic Usage

        >>> from obspy import read_inventory
        >>> resp = read_inventory()[0][0][0].response
        >>> resp.plot(0.001, output="VEL")  # doctest: +SKIP

        .. plot::

            from obspy import read_inventory
            resp = read_inventory()[0][0][0].response
            resp.plot(0.001, output="VEL")
        """
        import matplotlib.pyplot as plt
        from matplotlib.transforms import blended_transform_factory

        # detect sampling rate from response stages
        if sampling_rate is None:
            for stage in self.response_stages[::-1]:
                if (stage.decimation_input_sample_rate is not None and
                        stage.decimation_factor is not None):
                    sampling_rate = (stage.decimation_input_sample_rate /
                                     stage.decimation_factor)
                    break
            else:
                msg = ("Failed to autodetect sampling rate of channel from "
                       "response stages. Please manually specify parameter "
                       "`sampling_rate`")
                raise Exception(msg)
        if sampling_rate == 0:
            msg = "Can not plot response for channel with sampling rate `0`."
            raise ZeroSamplingRate(msg)

        t_samp = 1.0 / sampling_rate
        nyquist = sampling_rate / 2.0
        nfft = int(sampling_rate / min_freq)

        cpx_response, freq = self.get_evalresp_response(
            t_samp=t_samp, nfft=nfft, output=output, start_stage=start_stage,
            end_stage=end_stage)

        if axes is not None:
            ax1, ax2 = axes
            fig = ax1.figure
        else:
            fig = plt.figure()
            ax1 = fig.add_subplot(211)
            ax2 = fig.add_subplot(212, sharex=ax1)

        label_kwarg = {}
        if label is not None:
            label_kwarg['label'] = label

        # plot amplitude response
        lw = 1.5
        lines = ax1.loglog(freq, abs(cpx_response), lw=lw, **label_kwarg)
        color = lines[0].get_color()
        if self.instrument_sensitivity:
            trans_above = blended_transform_factory(ax1.transData,
                                                    ax1.transAxes)
            trans_right = blended_transform_factory(ax1.transAxes,
                                                    ax1.transData)
            arrowprops = dict(
                arrowstyle="wedge,tail_width=1.4,shrink_factor=0.8", fc=color)
            bbox = dict(boxstyle="round", fc="w")
            if self.instrument_sensitivity.frequency:
                ax1.annotate("%.1g" % self.instrument_sensitivity.frequency,
                             (self.instrument_sensitivity.frequency, 1.0),
                             xytext=(self.instrument_sensitivity.frequency,
                                     1.1),
                             xycoords=trans_above, textcoords=trans_above,
                             ha="center", va="bottom",
                             arrowprops=arrowprops, bbox=bbox)
            if self.instrument_sensitivity.value:
                ax1.annotate("%.1e" % self.instrument_sensitivity.value,
                             (1.0, self.instrument_sensitivity.value),
                             xytext=(1.05, self.instrument_sensitivity.value),
                             xycoords=trans_right, textcoords=trans_right,
                             ha="left", va="center",
                             arrowprops=arrowprops, bbox=bbox)

        # plot phase response
        phase = np.angle(cpx_response, deg=plot_degrees)
        if unwrap_phase and not plot_degrees:
            phase = np.unwrap(phase)
        ax2.semilogx(freq, phase, color=color, lw=lw)

        # plot nyquist frequency
        for ax in (ax1, ax2):
            ax.axvline(nyquist, ls="--", color=color, lw=lw)

        # only do adjustments if we initialized the figure in here
        if not axes:
            _adjust_bode_plot_figure(fig, show=False,
                                     plot_degrees=plot_degrees)

        if outfile:
            fig.savefig(outfile)
        else:
            if show:
                plt.show()

        return fig

    def get_paz(self):
        """
        Get Poles and Zeros stage.

        Prints a warning if more than one poles and zeros stage is found.
        Raises if no poles and zeros stage is found.

        :rtype: :class:`PolesZerosResponseStage`
        :returns: Poles and Zeros response stage.
        """
        paz = [deepcopy(stage) for stage in self.response_stages
               if isinstance(stage, PolesZerosResponseStage)]
        if len(paz) == 0:
            msg = "No PolesZerosResponseStage found."
            raise Exception(msg)
        elif len(paz) > 1:
            msg = ("More than one PolesZerosResponseStage encountered. "
                   "Returning first one found.")
            warnings.warn(msg)
        return paz[0]

    def get_sacpz(self):
        """
        Returns SACPZ ASCII text representation of Response.

        :rtype: str
        :returns: Textual SACPZ representation of response.
        """
        # extract paz
        paz = self.get_paz()
        return paz_to_sacpz_string(paz, self.instrument_sensitivity)

    @classmethod
    def from_paz(cls, zeros, poles, stage_gain,
                 stage_gain_frequency=1.0, input_units='M/S',
                 output_units='VOLTS', normalization_frequency=1.0,
                 pz_transfer_function_type='LAPLACE (RADIANS/SECOND)',
                 normalization_factor=1.0):
        """
        Convert poles and zeros lists into a single-stage response.

        Takes in lists of complex poles and zeros and returns a Response with
        those values defining its only stage. Most of the optional parameters
        defined here are from
        :class:`~obspy.core.inventory.response.PolesZerosResponseStage`.

        :type zeros: list[complex]
        :param zeros: All zeros of the response to be defined.
        :type poles: list[complex]
        :param poles: All poles of the response to be defined.
        :type stage_gain: float
        :param stage_gain: The gain value of the response [sensitivity]
        :returns: new Response instance with given P-Z values
        """
        pzstage = PolesZerosResponseStage(
            stage_sequence_number=1, stage_gain=stage_gain,
            stage_gain_frequency=stage_gain_frequency,
            input_units=input_units, output_units=output_units,
            pz_transfer_function_type=pz_transfer_function_type,
            normalization_frequency=normalization_frequency,
            zeros=zeros, poles=poles,
            normalization_factor=normalization_factor)
        sens = InstrumentSensitivity(value=stage_gain,
                                     frequency=stage_gain_frequency,
                                     input_units=input_units,
                                     output_units=output_units)
        resp = cls(instrument_sensitivity=sens, response_stages=[pzstage])
        resp.recalculate_overall_sensitivity()
        return resp


def paz_to_sacpz_string(paz, instrument_sensitivity):
    """
    Returns SACPZ ASCII text representation of Response.

    :type paz: :class:`PolesZerosResponseStage`
    :param paz: Poles and Zeros response information
    :type instrument_sensitivity: :class:`InstrumentSensitivity`
    :param paz: Overall instrument sensitivity of response
    :rtype: str
    :returns: Textual SACPZ representation of poles and zeros response stage.
    """
    # assemble output string
    out = []
    out.append("ZEROS %i" % len(paz.zeros))
    for c in paz.zeros:
        out.append(" %+.6e %+.6e" % (c.real, c.imag))
    out.append("POLES %i" % len(paz.poles))
    for c in paz.poles:
        out.append(" %+.6e %+.6e" % (c.real, c.imag))
    constant = paz.normalization_factor * instrument_sensitivity.value
    out.append("CONSTANT %.6e" % constant)
    return "\n".join(out)


class InstrumentSensitivity(ComparingObject):
    """
    From the StationXML Definition:
        The total sensitivity for a channel, representing the complete
        acquisition system expressed as a scalar. Equivalent to SEED stage 0
        gain with (blockette 58) with the ability to specify a frequency range.

    Sensitivity and frequency ranges. The FrequencyRangeGroup is an optional
    construct that defines a pass band in Hertz (FrequencyStart and
    FrequencyEnd) in which the SensitivityValue is valid within the number of
    decibels specified in FrequencyDBVariation.
    """
    def __init__(self, value, frequency, input_units,
                 output_units, input_units_description=None,
                 output_units_description=None, frequency_range_start=None,
                 frequency_range_end=None, frequency_range_db_variation=None):
        """
        :type value: float
        :param value: Complex type for sensitivity and frequency ranges.
            This complex type can be used to represent both overall
            sensitivities and individual stage gains. The FrequencyRangeGroup
            is an optional construct that defines a pass band in Hertz (
            FrequencyStart and FrequencyEnd) in which the SensitivityValue is
            valid within the number of decibels specified in
            FrequencyDBVariation.
        :type frequency: float
        :param frequency: Complex type for sensitivity and frequency
            ranges.  This complex type can be used to represent both overall
            sensitivities and individual stage gains. The FrequencyRangeGroup
            is an optional construct that defines a pass band in Hertz (
            FrequencyStart and FrequencyEnd) in which the SensitivityValue is
            valid within the number of decibels specified in
            FrequencyDBVariation.
        :param input_units: string
        :param input_units: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Name of units, e.g. "M/S", "V", "PA".
        :param input_units_description: string, optional
        :param input_units_description: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :param output_units: string
        :param output_units: The units of the data as output from the
            perspective of data acquisition. These would be the units of the
            data prior to correcting for this response.
            Name of units, e.g. "M/S", "V", "PA".
        :type output_units_description: str, optional
        :param output_units_description: The units of the data as output from
            the perspective of data acquisition. These would be the units of
            the data prior to correcting for this response.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :type frequency_range_start: float, optional
        :param frequency_range_start: Start of the frequency range for which
            the SensitivityValue is valid within the dB variation specified.
        :type frequency_range_end: float, optional
        :param frequency_range_end: End of the frequency range for which the
            SensitivityValue is valid within the dB variation specified.
        :type frequency_range_db_variation: float, optional
        :param frequency_range_db_variation: Variation in decibels within the
            specified range.
        """
        self.value = value
        self.frequency = frequency
        self.input_units = input_units
        self.input_units_description = input_units_description
        self.output_units = output_units
        self.output_units_description = output_units_description
        self.frequency_range_start = frequency_range_start
        self.frequency_range_end = frequency_range_end
        self.frequency_range_db_variation = frequency_range_db_variation

    def __str__(self):
        ret = ("Instrument Sensitivity:\n"
               "\tValue: {value}\n"
               "\tFrequency: {frequency}\n"
               "\tInput units: {input_units}\n"
               "\tInput units description: {input_units_description}\n"
               "\tOutput units: {output_units}\n"
               "\tOutput units description: {output_units_description}\n")
        ret = ret.format(**self.__dict__)
        return ret

    def _repr_pretty_(self, p, cycle):
        p.text(str(self))


# XXX duplicated code, PolynomialResponseStage could probably be implemented by
# XXX inheriting from both InstrumentPolynomial and ResponseStage
class InstrumentPolynomial(ComparingObject):
    """
    From the StationXML Definition:
        The total sensitivity for a channel, representing the complete
        acquisition system expressed as a polynomial. Equivalent to SEED stage
        0 polynomial (blockette 62).
    """
    def __init__(self, input_units, output_units,
                 frequency_lower_bound,
                 frequency_upper_bound, approximation_lower_bound,
                 approximation_upper_bound, maximum_error, coefficients,
                 approximation_type='MACLAURIN', resource_id=None, name=None,
                 input_units_description=None,
                 output_units_description=None, description=None):
        """
        :type approximation_type: str
        :param approximation_type: Approximation type. Currently restricted to
            'MACLAURIN' by StationXML definition.
        :type frequency_lower_bound: float
        :param frequency_lower_bound: Lower frequency bound.
        :type frequency_upper_bound: float
        :param frequency_upper_bound: Upper frequency bound.
        :type approximation_lower_bound: float
        :param approximation_lower_bound: Lower bound of approximation.
        :type approximation_upper_bound: float
        :param approximation_upper_bound: Upper bound of approximation.
        :type maximum_error: float
        :param maximum_error: Maximum error.
        :type coefficients: list[float]
        :param coefficients: List of polynomial coefficients.
        :param input_units: string
        :param input_units: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Name of units, e.g. "M/S", "V", "PA".
        :param output_units: string
        :param output_units: The units of the data as output from the
            perspective of data acquisition. These would be the units of the
            data prior to correcting for this response.
            Name of units, e.g. "M/S", "V", "PA".
        :type resource_id: str
        :param resource_id: This field contains a string that should serve as a
            unique resource identifier. This identifier can be interpreted
            differently depending on the data center/software that generated
            the document. Also, we recommend to use something like
            GENERATOR:Meaningful ID. As a common behavior equipment with the
            same ID should contains the same information/be derived from the
            same base instruments.
        :type name: str
        :param name: A name given to the filter stage.
        :param input_units_description: string, optional
        :param input_units_description: The units of the data as input from the
            perspective of data acquisition. After correcting data for this
            response, these would be the resulting units.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :type output_units_description: str, optional
        :param output_units_description: The units of the data as output from
            the perspective of data acquisition. These would be the units of
            the data prior to correcting for this response.
            Description of units, e.g. "Velocity in meters per second",
            "Volts", "Pascals".
        :type description: str, optional
        :param description: A short description of of the filter.
        """
        self.input_units = input_units
        self.output_units = output_units
        self.input_units_description = input_units_description
        self.output_units_description = output_units_description
        self.resource_id = resource_id
        self.name = name
        self.description = description
        self._approximation_type = approximation_type
        self.frequency_lower_bound = frequency_lower_bound
        self.frequency_upper_bound = frequency_upper_bound
        self.approximation_lower_bound = approximation_lower_bound
        self.approximation_upper_bound = approximation_upper_bound
        self.maximum_error = maximum_error
        self.coefficients = coefficients

    @property
    def approximation_type(self):
        return self._approximation_type

    @approximation_type.setter
    def approximation_type(self, value):
        value = str(value).upper()
        allowed = ("MACLAURIN",)
        if value not in allowed:
            msg = ("Value '%s' for polynomial response approximation type not "
                   "allowed. Possible values are: '%s'")
            msg = msg % (value, "', '".join(allowed))
            raise ValueError(msg)
        self._approximation_type = value


class FilterCoefficient(FloatWithUncertainties):
    """
    A filter coefficient.
    """
    def __init__(self, value, number=None):
        """
        :type value: float
        :param value: The actual value of the coefficient
        :type number: int, optional
        :param number: Number to indicate the position of the coefficient.
        """
        super(FilterCoefficient, self).__init__(value)
        self.number = number

    @property
    def number(self):
        return self._number

    @number.setter
    def number(self, value):
        if value is not None:
            value = int(value)
        self._number = value


class CoefficientWithUncertainties(FloatWithUncertainties):
    """
    A coefficient with optional uncertainties.
    """
    def __init__(self, value, number=None, lower_uncertainty=None,
                 upper_uncertainty=None):
        """
        :type value: float
        :param value: The actual value of the coefficient
        :type number: int, optional
        :param number: Number to indicate the position of the coefficient.
        :type lower_uncertainty: float
        :param lower_uncertainty: Lower uncertainty (aka minusError)
        :type upper_uncertainty: float
        :param upper_uncertainty: Upper uncertainty (aka plusError)
        """
        super(CoefficientWithUncertainties, self).__init__(
            value, lower_uncertainty=lower_uncertainty,
            upper_uncertainty=upper_uncertainty)
        self.number = number

    @property
    def number(self):
        return self._number

    @number.setter
    def number(self, value):
        if value is not None:
            value = int(value)
        self._number = value


def _adjust_bode_plot_figure(fig, plot_degrees=False, grid=True, show=True):
    """
    Helper function to do final adjustments to Bode plot figure.
    """
    import matplotlib.pyplot as plt
    # make more room in between subplots for the ylabel of right plot
    fig.subplots_adjust(hspace=0.02, top=0.87, right=0.82)
    ax1, ax2 = fig.axes[:2]
    # workaround for older matplotlib versions
    try:
        ax1.legend(loc="lower center", ncol=3, fontsize='small')
    except TypeError:
        leg_ = ax1.legend(loc="lower center", ncol=3)
        leg_.prop.set_size("small")
    plt.setp(ax1.get_xticklabels(), visible=False)
    ax1.set_ylabel('Amplitude')
    minmax1 = ax1.get_ylim()
    ax1.set_ylim(top=minmax1[1] * 5)
    ax1.grid(True)
    ax2.set_xlabel('Frequency [Hz]')
    if plot_degrees:
        # degrees bode plot
        ax2.set_ylabel('Phase [degrees]')
        ax2.set_yticks(np.arange(-180, 180, 30))
        ax2.set_yticklabels(np.arange(-180, 180, 30))
        ax2.set_ylim(-180, 180)
    else:
        # radian bode plot
        plt.setp(ax2.get_yticklabels()[-1], visible=False)
        ax2.set_ylabel('Phase [rad]')
        minmax2 = ax2.yaxis.get_data_interval()
        yticks2 = np.arange(minmax2[0] - minmax2[0] % (pi / 2),
                            minmax2[1] - minmax2[1] % (pi / 2) + pi, pi / 2)
        ax2.set_yticks(yticks2)
        ax2.set_yticklabels([_pitick2latex(x) for x in yticks2])
    ax2.grid(True)

    if show:
        plt.show()


def _pitick2latex(x):
    """
    Helper function to convert a float that is a multiple of pi/2
    to a latex string.
    """
    # safety check, if no multiple of pi/2 return normal representation
    if x % (pi / 2) != 0:
        return "%#.3g" % x
    string = "$"
    if x < 0:
        string += "-"
    if x / pi % 1 == 0:
        x = abs(int(x / pi))
        if x == 0:
            return "$0$"
        elif x == 1:
            x = ""
        string += r"%s\pi$" % x
    else:
        x = abs(int(2 * x / pi))
        if x == 1:
            x = ""
        string += r"\frac{%s\pi}{2}$" % x
    return string


if __name__ == '__main__':
    import doctest
    doctest.testmod(exclude_empty=True)
